UV-Vis-NIR spectroscopy is routinely used in analytical chemistry for the quantitative determination of various analytes, such as transition metal ions, organic compounds, and biological macromolecules.
- Reviews (0)
|Testing Method||Ultraviolet-visible-near infrared spectroscopy (UV-Vis-NIR)|
Ultraviolet-visible-near infrared spectroscopy (UV-Vis-NIR) refers to absorption or reflectance spectroscopy in the visible (400 – 700nm) and adjacent UV (200 – 400nm) and near-infrared (700 – 2500 nm) ranges of electromagnetic spectrum that causes many organic molecules to undergo electronic transitions.
The Beer-Lambert law states that the absorbance of a solution is directly proportional to the concentration of the absorbing species in the solution and the path length. Thus, for a fixed path length, UV-Vis-NIR spectroscopy can be used to determine the concentration of the absorber in a solution.
The wavelengths of absorption peaks can be correlated with the types of bonds in a given molecule and are valuable in determining the functional groups within a molecule.
UV-Vis-NIR spectroscopy is routinely used in analytical chemistry for the quantitative determination of different analytes, such as transition metal ions, highly conjugated organic compounds, and biological macromolecules. Spectroscopic analysis is commonly carried out in solutions but solids may also be studied.
(1) Determine concentrations of an absorbing species in solution in a quantitative way; the absorbing species can be transition metal ions, organic compounds, or biological macromolecules.
(2) Detect impurities in an organic liquid.
(3) Determine the kinetics or rate constant of a chemical reaction.
(4) Measure thickness and optical properties of thin film on a wafer.
(5) Applications of NIR spectroscopy include the analysis of foodstuffs, pharmaceuticals, combustion products, and a major branch of astronomical spectroscopy.
Wikipedia: Ultraviolet–Visible Spectroscopy
Wikipedia: Near-infrared Spectroscopy
High Performance Liquid Chromatography (HPLC)
High performance liquid chromatography (HPLC) is a technique in analytical chemistry used to separate, identify, and quantify each component in a mixture.
Gas Chromatography – Mass Spectrometry (GC-MS)
Gas chromatography – mass spectrometry (GC-MS) combines the features of gas chromatography and mass spectrometry to identify different substances within a test sample.
Mass Spectrometry (MALDI-TOF-MS)
Mass spectrometry (MALDI-TOF-MS) is an analytical technique that ionizes chemical species and sorts the ions based on their mass to charge ratio.
High-speed centrifuge is a method widely used to separate two immiscible substances involving the application of the centripetal force.
Gas Chromatography with FID/TCD/MS Detectors
Gas chromatography (GC) is used in analytical chemistry for separating and analyzing compounds that can be vaporized without decomposition.
Brunauer-Emmett-Teller Analyzer (BET)
Brunauer-Emmett-Teller analyzer is the most common method for determining the surface area and pore size distribution of powders and porous materials.
Mass Spectrometry (Q-TOF-MS)
Mass spectrometry (Q-TOF-MS) is an analytical technique that ionizes chemical species and sorts the ions based on their mass to charge ratio.
High Performance Liquid Chromatography – Mass Spectrometry (HPLC-MS)
High performance liquid chromatography – mass spectrometry (HPLC-MS) combines the physical separation capabilities of liquid chromatography with the mass analysis capabilities of mass spectrometry.
Fluorescence spectroscopy measures the fluorescent light emitted from a sample at different wavelengths, after illumination with a xenon flash lamp.
Laser Light Scattering (LLS)
Laser light scattering (LLS) is used to determine size of various particles including proteins, polymers, micelles and nanoparticles.